Patterson Dental was founded in the year 1877 by Dave G. Misiak and has grown into one of the largest distributors of dental merchandise, with Patterson Dental handpieces ranking exceptionally high.
In addition to the dental handpieces, the company offers over 100,000 different products, is considered to be experts in dental technology, and provide a multitude of solutions and services for the dental practitioner.
Today, Patterson Dental handpieces can be found across in the nation, in the offices of many practitioners.
If you are reading this, you likely utilize these handpieces in your practice. In this brief guide, we will be reviewing the most common maintenance mistakes made with Patterson Dental handpieces.
“If you don’t understand how to run an efficient operation, new machinery will just give you new problems of operation and maintenance. The sure way to increase productivity is to better administrate man and machine…” – W. Edwards Deming
Common Maintenance Mistakes
Patterson Dental handpieces offer practitioners a high level of durability, stability, and quality; however, when dental practices elect to perform their own maintenance on these handpieces, they may – unknowingly – make many mistakes. These include the following:
- One of the first – and, typically the most common – of all mistakes is that many practitioners utilize a chemical cleansing agent to wipe down the Patterson Dental handpieces in their office prior to engaging in the process of sterilization. It is a known fact that these cleansing agents leave behind a residue consisting of synthetic substances. This residue could result in reactions that may harm the dental handpiece instrument when the heat from the sterilization process comes in contact with the handpiece.
- Many practitioners fail to lubricate the appropriate hole on their Patterson Dental handpiece when attempting to maintain their instrument. Lubrication should only be placed on the air line hole of the unit. This way, the lubrication is directed to the turbine unit. If lubrication is placed anywhere else, lack of functionality and inoperability could – and likely will – occur.
- Prior to autoclaving, many practitioners neglect to allow their Patterson Dental handpiece to run. This is a step that should be completed immediately after the lubrication process. If your handpiece is not allowed to run prior to autoclaving, it is quite likely that the lubrication that you used will turn into a thick, gum-like substance that will have a detrimental impact on the turbine of the piece. Additionally, it is likely to result in discomfort among your patients during procedures as lubrication may spew out of the instrument.
- The autoclaves of the Patterson Dental handpieces should be properly maintained. Over time, debris will accumulate in the unit. In turn, this debris will completely contaminate your dental handpieces. Furthermore, it has the capability of contaminating the lines and the evacuation pieces associated with your Patterson Dental handpieces.
As you can see, many mistakes may be made when attempting to maintain your Patterson Dental handpieces.
If you want to avoid these issues, you should hire a certified dental handpiece repair and maintenance technician. These individuals are skilled in caring for an assortment of dental instruments, provide top-of-the-line customer service, and highly competitive prices.